This multipurpose liquid flow meter uses a turbine mechanism to give accurate flow measurements of cold water. It’s constructed of durable plastic which makes it economical and chemical resistant. This versatile flow meter has a resettable batch total, flow rate and it also offers a lifetime total! In addition, its’ ability to be calibrated in the field allows you to make adjustments to fine tune your measurements. Need to measure in something other than gallons? No problem for the DM Series! It can report data in Gallons, Liters, Pints and Quarts. For ease of reading, the display can also be rotated to any of the 4 possible positions (see photos for 2 different display positions).
- 1/4", 1/2", 3/4" or 1" NPT Inlet and Outlet
- +/- 1% Full Scale
- Max Working Pressure 145 PSI
- Flow Rate Ranges (NOTE: WHEN RUNNING AT VERY LOW FLOWS, VERTICAL MOUNT IS RECOMMENDED)
- 1/4" - 1.32-105 GPH (0.022- 1.75 GPM)
- 1/2" - 2.6-395 GPH (0.04333 - 6.583 GPM)
- 3/4" - 0.22 to 22 GPM
- 1" - 0.35 to 35 GPM
- (See manual for more details)
- Displays up to 99,999
- Meter comes pre-calibrated for water applications
- Single Measurement Reset
- Offers Lifetime Total
- Flow Rate
- Can be calibrated in the field
- Multiple Measurement Units
- Replaceable Battery
- Rotatable Display for Ease of Reading
- Max Temp 140°F
- Continuous Screen Display
- Powered by 2 AAA batteries (included)
- Garden Hose Water Measurement (with optional garden hose adapters)
- Fertilizer Trucks
- Plant Process Water
- Tank Filling
- Cement Mixers
- Pond Water Feed Lines
- Chemical Feed Lines
Enclosure: Pa66 + GF
Rotor: Pa66 / POM / PPS
Shaft: stainless steel 316
- Display Square: 3" x 3" (including rubber fins)
- Inlet to outlet: Approx 3.75"
- LCD Screen: 1 3/4" X 1 1/8"
13 Reviews Hide Reviews Show Reviews
I bought this to measure the flow rate on an open looped geo thermal HVAC unit. Just as I suspected, the flow rate was at 14 GPM. The meter allowed me to reduce the flow rate to 6 GPM where it needs to be. Now I am saving 8 GPM which reduces the work my well pump has to do. Good product. It easily connected to the outflow PVC pipe.
The threaded connections are real tight, however, the plastic material may result is cross-treading the adapters. Overall, it is performing well for us.
I purchased this unit (1/2” version) to measure drip watering for my plants so that I can measure how much liquid fertilizer to add to the water. I was not successful. May be because I purchased a unit with an incorrect measurement range. Sometimes it measures 0.1 pints per minute sometimes 0.2 pints, most of the time it stays at 0.0. So I don’t know if it works, I would like to think it does, especially since it is $150 item. I reached out to customer service, they were polite, but in the end the conversation did end in resolution of my problem and I still don’t know how much water is flowing. Anyway thank you.
RESPONSE FROM FLOWS.COM: We’re sorry this product didn’t work for your application. As you mentioned, it was measuring 0.1 to 0.2 pints/minute at times. When you convert that to Gallons, you’re looking at 0.0125 to 0.025 Gallons per minute. The ½” meter in this series is only going to be accurate between 0.04333-6.583 GPM. When purchasing any meter the most important part of the selection process is to make sure your flow rate meets that meter’s specs. This usually can be done with a “bucket” test (time how long a known amount of water takes to dispense and calculate that out to Gallons Per Minute). If the meter had been defective, it does have a 1 year warranty that we would honor. In addition, if we had a meter that measures that low, we would have been happy to offer a discount on the purchase of another meter.
Leaks, made really cheap for the price.
Response from Flows.Com:
We're sorry you're having an issue with this product. We reached out to you two times to discuss as it does have a 1 year warranty on any defects. Leaking would likely be a valid warranty issue (if it wasn't caused by something outside of what is on the meter's specification sheet ie: max pressure, temperature, etc.).
There is no on/off switch on this. You have to take it apart and remove the batteries each time. Poor engineering.
I recently purchased 2 DM-P-050 flow meters to monitor the flow of fertilizer into my irrigation system. I love the ease of use and the precision. It requires a bit of recalibration since I am using it for fertilizer and not water. Very helpful that the orientation can be changed so I can run them vertical for low flow operations.
I bought this for a low yield well in order to adjust the pumping rate of the well. Simple to install and it has been working perfectly for a week. It has been invaluable in setting the flow restriction valve to reach the desired flow rate. It has also been eye opening to see how much water I am using to irrigate a one-acre yard in the Central Valley of California! I highly recommend it.
I needed to drain my pool. This is a great investment! That's the good news... however, be aware that there is no power button... I assume this is because design is for permanent installation. Also display orientation can only be adjusted by removing four screws, removing display and reorienting prior to screwing it down. Either of these design considerations are not a problem for me.... and I don't want to remove the display just to unplug the batteries... they say batteries will last two years.